Edan M8/M9 Adult Finger Clip SpO2 Sensor, 6pin, 2.8m
|Compatible Model||BM3, M3A, M3B, M8, M8A, M8B, M9, M9A, M50, M80, IM8, IM8A, IM8B,
IM9, IM9A, IM50, IM80|
|Sensor Type||Adult/Child/Infant/Neonate/Multi Y/Ear clip/Animal|
|Probe House||ABS Shell ( Finger Clip)|
|Cable Material||TPU Light Gray Jacket, 4.0mm Diameter||Sensor Connector||gold pin|
|Cable Length||2.7M||Package||1 pc in a opp bag|
1. Hold the sensor with its opening towards the patient's index finger, toes, palm or sole. The side with a finger or toes
sign is placed on the top, detail is as follows:
2. Put the patient's measurement part into the sensor until the tip against at the end
of the sensor.
3. Adjust the measurement part to be placed evenly on the middle
base of the sensor.
4. Direct the cable along the top of the patient's hand or foot.
5. Plug the sensor into the proper oximeter and verify operation as
described in the oximeter operator's manual.
6. Inspect the sensor site every 2 hours for skin integrity.
1.0 The usage range
This specification suited for customer's appointed models.
2.0 Usage condition(GB9706.1-2007 10.2.1)
2.1 Rated temperature:25ºC.
2.2 Operation environment:10ºC~40ºC.
2.3 Storage and transport method:individual moistureproof package
for each Product,Transport under the conditions of ambient
3. Appearance requests:
3.1 a The cable is round and glossy,the color uniformity, the
jacket is ok,no Disrepair,no dirt,no disfigurement,etc.
3.1 b The surface of the connector is glossy, no burr,no obvious
disfigurement,no modification,no dirt,no metal rust etc.
3.1 c (EC13:2002 220.127.116.11/IEC60601-2-47 6.1aa)Clear dot-color and
characters, thecorresponding printing should be correct.(EC13:2002
3.1 d Product labels for use in line with the requirements of
3.1 e Product labels and graphics in line with the engraved die
EN980-2008 / EU CE requirementf Product cleaning, disinfection and
chemical resistance in ine with the requirements of EC53-1995 4.3
3.2 Size requirements:
To check the Size according to the drawing which was issued by the
engineer. (The important size requirements, please notify it on
3.3 Packaging requirements:
The packaging in accordance with the requirements of the engineer
drawin(Incase of special packaging requirements, please promptly
4.0 Technical requirements
ECG executive standards:ANSI/AAMI EC53:1995/(R)2001 Note:ANSI/AAMI
EC53:1995/(R)2001 Standard the not description
general requested by the executive.
4.2 Customer special electrical and mechanical performance
(Please check the Implementation standards in remarks.)
1. Please let us know the product model you need
2. We will make proforma invoice for you.
3. Checking and confirm the invoice.
4. After payment, all goods will be sent to you in 5-7 days as
5. OEM/ODM is welcomed.
T/T, Western Union.
1. EMS. DHL. UPS. FedEx, TNT.
2. It will take 3~7days for shipping (depend on country distance).
3. Any import charges or fees are buyer's responsibility.
4. We will provide you the tracking number once the goods sent,
tracking it until you get the package.
5. The shipping cost will be on clients' side.
We are Professional OEM/ODM Medical Cable Manufacturer,please feel free to
Cell phone:+86 17347272725